COVID-19 IgM Antibody Test Kit
Diagnostic Kit for IgM Antibody to 2019-nCoV(Colloidal Gold)
Catalog No.: BG1901C
5 test cards/ box, 20 test cards/ box, 40 test cards/ box
The kit is used for the qualitative detection of novel coronavirus (2019-nCoV) IgM antibodies in human serum and plasma samples in vitro. It is only used as a supplementary detection index for suspected negative cases of new coronavirus nucleic acid detection, or in collaboration with nucleic acid detection in the diagnosis of suspected cases, and can not be used as a basis for the diagnosis and elimination of pneumonia caused by new coronavirus infection.
The test results are positive and need further confirmation. Negative results can not exclude the possibility of infection.
1.Pretest preparation: 10μL, 50μL and 100μL micropipette 1 and matched suction heads or dropper (10μL, 50μL, 100μL).
Inspection process: Put the detection card on the dry horizontal worktable. Add 10μL serum or plasma sample into the sample well using the micro pipette, then add two drops(about70~100μL) of sample dilution immediately. Observe the result in 10~15 minutes after the serum or plasma samples added.
3. Precautions in the inspection process:
a.The kit and the samples to be tested should be placed in equilibrium at room temperature before they can be inspected.
b.Use the test card within 30 minutes once opened
c. The observation is invalid after 15 minutes.
【Interpretation of test results】
To develop color on C line only: negative;
To develop color on both C line and T line: positive;
To develop color on T line only: invalid;
If no C line is developed, the assay is invalid regardless of any color in the T line.
Note: Samples should be processed according to infectious contaminants and samples should be collected again.